oref1 (super advanced features)¶
NOTE OF CAUTION:
- oref1 is different than oref0, the baseline “traditional” OpenAPS implementation that only uses temporary basal rates.
- You should have run oref0 (basic OpenAPS looping) for more than two weeks, and be very aware of all the types of situations in which your rig might fail, before you enable oref1-related features.
- If running more than one rig, you will want to make sure all rigs are running an SMB-aware oref0 version (release 0.5.1 or higher) before enabling SMB on any of them (even if SMB is not enacted on all rigs, all rigs need to know about it).
Only run oref1 with the following caveats in mind:¶
- Remember that you are choosing to test a still-in-development feature. Do so at your own risk & with due diligence to keep yourself safe. The number one rule (and the first pass phrase for enabling oref1 features) is “s@fety”.
- You should have run oref0 (basic OpenAPS looping) for more than two weeks, and be very aware of all the types of situations in which your rig might fail.
- We are requiring that you also have run autotune prior to enabling SMB. Why? Because if you have wonky ISF settings, for example, you may be more likely to go low or high with SMB. It will help a lot to have run autotune and be aware if the algorithm is recommending changes to ISF, basal, and/or carb ratio. You are not required to run autotune automatically/nightly as part of your loop with SMB; but you should at least run it manually and get an idea for how confident you are in your settings being right or not; and keep that in mind when evaluating SMB outcomes for yourself.
- You should have basals of > 0.5 U/hr. (SMB is not advisable for those with very small basals; since 0.1U is the smallest increment that can be bolused by SMB. We also added a basal check to disable SMB when basals are < 0.3 U/hr. If your “regular” basal in the pump is 0.3 U/hr and autosens or autotune has adjusted your basal rate to below 0.3 U/hr, SMBs will be disabled as well.)
- Read the following:
- A. The updated reference design (https://openaps.org/reference-design/) that explains the differences between oref0 and oref1
- B. The following two posts for background on oref1:
- Make sure you understand what SMB & UAM stand for (read the above posts, we know you skipped them!)
- Plan to have a learning curve. You will interact with oref1 differently when on SMB and UAM than how you were interacting with oref0. In particular: do not do correction boluses based on insulinReq; instead use temp targets to give the rig a “nudge”. You are very likely to overshoot if you try to do things manually on top of what SMB has already done!
- SMB may not be for everyone. Like everything else, plan to test it, fall back to previous methods of diabetes treatment if needed, and give yourself a time period for deciding whether or not it works well for you.
SMB, like all things in OpenAPS, is designed with safety in mind. (Did you skip reading the updated reference design? Go read that first!) SMB is designed to give you reasonably SAFE amounts of bolus needed upfront and use reduced temporary basal rates to safely balance out the peak insulin timing. You are likely to see many long low or zero temps (upwards of 120 minutes long) with SMB turned on, while oref1 is administering SMBs or waiting until it’s safe to do so.
Single SMB amounts are limited by several factors. The largest a single SMB bolus can be is the SMALLEST value of:
- 30 minutes of the current regular basal rate (as adjusted by autotune/autosens), or
- 1/3 of the Insulin Required amount, or
- the remaining portion of your maxIOB setting in preferences
(History of SMB development: https://github.com/openaps/oref0/issues/262)
UAM will be triggered if the preference is toggled on and there is carb activity detected based on positive deviations.
(History of UAM development: https://github.com/openaps/oref0/issues/297)
How to turn on SMB/UAM¶
- As of July 13, 2017, SMB/UAM are in the master branch of oref0 (oref0 0.5.0). They are under the advanced features menu, and require you to read these docs carefully from top to bottom on this page in order to enable it during the setup script. Afterward, you also need to turn on the relevant settings in preferences.json. You may want to experiment with turning only one of them on at a time so you can closely observe the behavior (via both Nightscout and pump-loop.log) in the enabled situation. In addition to testing oref1 in “normal” situations, pay special attention to how it behaves in more extreme situations, such as with rescue carbs (announced or not), post-meal activity, etc.
There are multiple preference toggles for SMB/UAM. Check out the preferences page for more details on all the settings, but the short version is:
* enableSMB_with_bolus means SMB will be enabled for the DIA (in hours) after a manual bolus * enableSMB_with_COB means SMB will be enabled when you've entered carbs * enableSMB_with_temptarget means SMB will be enabled with eating soon or lower temp targets. For example, if your target is usually 100mg/dL, a temp target of 99 (or 80, the typical eating soon target) will enable SMB. Conversely, a higher temp target (101 if your target is 100) will disable SMB.
- To test UAM, you’ll need to toggle to “true” the enableUAM in preferences.json. UAM can be enabled without SMB, but it won’t be very effective without enableSMB_with_bolus. You’ll probably also want to make sure your nightscout is updated to the latest version to make sure that you can take advantage of UAM prediction lines. To test UAM, you’ll first want to be familiar with SMB’s “normal” behavior (related, the second safety phrase you need to know is “gate”), and then test, with a small meal, giving an up-front bolus (of more than 30m worth of basal, so it can be distinguished from an SMB) and not entering carbs. You’ll probably also want to do an “eating soon mode” temporary target or bolus beforehand. You can then observe, by watching the NS purple line predictions and the pump-loop.log, whether your OpenAPS rig is SMB’ing appropriately when it starts to see UAM carb impact.
- Make sure you read the above, especially the “only enable oref1 if...” section. SMB will behave differently than oref0 would. Watch carefully, and use your common sense and do what’s right for you & your diabetes.
- Common errors include:
- Not changing the preferences to be “true” for the relevant settings after you’ve enabled the oref1 features. You should see “Starting supermicrobolus pump-loop at...” in pump-loop.log if you have successfully enabled everything.
- Not running autotune. Remember, you don’t have to enable it to run as part of your loop at night, but you should run it manually, review the results, and otherwise be VERY confident in your underlying pump settings (basals, ISF, carb ratio) before using oref1.
- Pump clock being >1 minute off from rig’s time. This means 60 seconds. Not 61 seconds; 68 seconds; 90 seconds. Needs to be less than 60 seconds apart.
"Checking pump clock: "2017-05-16T15:46:32-04:00" is within 1m of current time: Tue May 16 15:47:40 EDT 2017is an example of a >60 second gap that needs fixing before it will work properly. We added a script to automatically attempt to fix the pump time in case of a >60 second difference, but you may occasionally see this type of error in the logs until the script is able to properly adjust the pump time.
Pushover, SMB, and OpenAPS¶
This is for OpenAPS-specific pushovers related to oref1 features about insulin required (insulinReq) and carbs required (carbsReq). Pushover is a way to easily send messages to your phone from another device with simple messages. If you have Pushover set up for Nightscout, you still need to tell your OpenAPS rig your Pushover information to get these rig-driven alerts. Nightscout Pushover alerts are separate and distinct from OpenAPS-generated Pushover alerts. Each can exists with or without the other.
If Pushover API token and User key were added during the setup script and you have oref1 enabled, you can get Pushover alerts in the following situations:
- When OpenAPS thinks carbs are needed to bring eventual BG up, and a 30m low temp won’t be enough to do it
- When SMB is active and hitting maxBolus. This is intended to alert you when SMB is going “all out”, and will tell you the total amount of insulin OpenAPS thinks you require (insulinReq) if current BG trends continue. DO NOT just blindly bolus for the amount of insulinReq. You will also see that the pushover alert lists the amount it is attempting to SMB. You should use this notification as a reminder to tell the rig about anything you know it doesn’t (like “oh yea, I want to enter my carbs for this meal”, or “oh, hold on, I need an activity mode, because I’m gonna go for a walk in a few minutes”). You can also decide if a manual meal bolus is appropriate, or if you’d like to manually bolus part of the insulinReq. If you’re just using insulinReq and not doing a normal meal bolus, you should NOT do the full insulinReq as a manual bolus, as oref1 is already attempting to deliver part of it as a SMB. SMB is designed to administer the insulinReq a little at a time, in order to be able to safely react if the BG rise slows or stops, so in cases where you might otherwise consider a correction bolus, it’ll often be best to not do anything at all and let SMB safely handle the increased need for insulin. If you do choose to do a small manual correction bolus for a portion of the insulinReq, be sure to subtract out the SMB oref1 is already delivering, and round down for safety.
- You are likely to cause yourself a low if you manually administer too much insulin. Be very careful about doing manual boluses based on Pushover alerts; see above about not doubling up on a microbolus that’s just been delivered.
- If the rig attempts to deliver a microbolus AND you have the bolus wizard menu open, it may cause the pump to error (and maybe reset). Recommendation: If you are getting Pushover alerts and decide to manually bolus in addition to the SMBs, you may want to use the “easy bolus” (up button arrow) method for bolusing, which is less likely to cause the pump to receive this error. When using the easy bolus, you may not be able to deliver the easy bolus if the rig has sent an SMB underneath. In that case, you’ll have to hit escape, wait for the SMB to finish delivering, and then perform your manual bolus (adjusting for the insulin just delivered).
If you are new to Pushover:¶
Pushover is a way to easily send messages to your phone from another device with simple messages. (kind of like getting a text message from your OpenAPS rig), but to use this you must first have Pushover installed on your iPhone or Android (download from your OS’s store).
- Log into https://pushover.net/. From this page you will see your User Key.
- At the bottom of the page you will see “Your Applications (Create an Application/API Token)”. You must first create an API Token:
- Click on the link provided. You must supply a name for your application, such as “OpenAPS”, and change the type to Script
- Then Check the box “By checking this box, you agree that you have read our Terms of Service and our Guide to Being Friendly to our API”
To put these in your setup you must add them to the oref0-setup.sh parameters, either by saying “Yes” to advanced features in the oref0-setup.sh script and entering the info there, or by using
edit-runagain and adding
--pushover_token=yourpushoverAPItoken --pushover_user=yourpushoveruserkey to the end of your runagain line. Then
bash ~/myopenaps/oref0-runagain.sh in order to rerun your script.